Atago 5014 Portable Refracto-Polarimeter RePo-4, Angle of Rotation : -5.00 to +5.00°(*) Brix : 0.0 to 85.0% - determine the amount of fructose (%) and honey moisture (%)
The RePo-4 can determine the amount of fructose (%) and honey moisture (%).
In addition to angle of rotation and Brix, the RePo-4 also converts and displays the amount of fructose (%) and honey moisture (%). The RePo-4 is ideal for total quality control of honey.
|Measurement Readings||Angle of Rotation,Brix,|
Temperature,Fructose% of honey, Honey Moisture %
|Range||Angle of Rotation:-5.00 to +5.00°(*)|
*Convertible to -25.00°to +25.00°with a 100mm observation tube
*Convertible to -50.00° to +50.00°with a 200mm observation tube
Brix: 0.0 to 85.0%
Fructose % : 0.0 to 99.9%
Honey Moisture % : 13.0 to 30.0%
Temperature : 15.0 to 40.0℃
|Resolution||Angle of Rotation : 0.01°|
Brix : 0.1%
Fructose % : 0.1%
Honey Moisture % : 0.1%
Temperature : 0.1℃
|Accuracy||Angle of Rotation : ±0.1° (at 20℃)|
Brix : ±0.2%
Honey Moisture % : ±0.2%
Temperature : ±1℃
|Temperature compensation||Brix ：15 to 40℃|
Honey Moisture % : 15 to 40℃
|Measurement Time||12 seconds|
|Power Supply||Size AAA alkaline battery x4|
|International Protection Class||IP67|
· Standard liquid Repo-series : RE-99110
RePo series Four points
Point 12 in 1
Brix alone can only indicate the overall amount of dissolved solids in a solution. With REPO-1, it is possible to determine if a product has been adulterated by measuring the optical rotation. For example, fructose rotates light to the left while glucose rotates light to the right. If a sample has been adulterated with glucose, it will make the optical rotation readings appear more positive than a pure solution.
Point 2Compact size
and weighs only 325g.
Point 3Automatically calculated scales
Press "User" to select the desired scale: Purity, International Sugar Scale, Specific Rotation, and Concentration.
Point 4Measurement evaluation indicator
Measurements are evaluated against pre-set tolerancelimits, and the result is indicated by green or red light.
(Upper and lower tolerance limits must be set in advance to using this function)
RePo series Features
Features and design that Atago's commitment created
RePo series Measuring Method
❶ Fill the sample stage up to the fill line (approx. 3mL).
❷ Press START.
❸ Measurement is displayed.